Capilair electroforesis dad detector 32 karat software manual
Director Global Marketing,. Insert the calibration disk for the attached sensor into the DVD drive ( the serial number for the sensor is written on the disk). Agilent 1100 HPLC G1315B Diode Array Detector. Always begin use at a reduced sensitivity level; increase sensitivity only.
PowerPac Basic Power Supply Instruction Manual For technical support, call your local Bio- Rad office, or in the U. The Agilent 1100 G1315B Diode Array Detector verifies separation quality with peak purity, multiple signals, and spectral libraries during analytical quantification, offering an extended wavelength range with the highest sensitivity. Agilent 1260 Infinity II Diode Array Detector HS ( G7117C) Analog output Recorder/ integrator: 100 mV or 1 V, output range 0. PhoneFax Sales and Technical Support China Varian Pan- Pacific Beijing Branch No.
This detector is for. The Agilent 1260 Infinity II Diode Array Detector HS offers full spectral detection for compound identification using spectral libraries or verification of the separation quality. Capilair electroforesis dad detector 32 karat software manual. 8 Yun Cheng Street. 08 or above LC and CE Drivers A. The Software CD also includes ViVA™ software which is the viewing application used to perform detector calibration, detector set- up, image.
14 or above For details about supported software versions refer to the compatibility matrix of your version of the LC and CE. Page 1 OWNER’ S MANUAL If you do not have prior experience with a metal detector, we strongly recommend that you: 1) Adjust the Sensitivity to a low setting in the event of false signals. The CD should start automatically. A complete list of all SFRs implemented by the ADC is provided in Table 8- 1.